Brava® Protective Ring
The Brava® Protective Ring was found to significantly reduce leakage compared to other rings on the market using a well-defined, validated test method*1.
Find out how we tested the Brava® Protective Ring
All rings were tested using a validated laboratory skin model. To ensure uniformity, all products were cut and pressed to the same size with a thickness of 2.0 mm. The quality of the skin was verified by measurement of transepidermal water loss.
A total of 10 replicates from 2 different lots of each product were applied to the skin surface. Real ileostomy output was introduced. The samples were incubated at 32°C and shaken at 80 rpm.
After 24 hours, the samples were peeled off the skin surface and the leakage area under the ring was measured. Brava® Protective Ring was found to significantly reduce leakage compared to all other rings tested*1.
Do you have more questions? Get in touch with us at 1-866-293-6349 or at email@example.com
*Study was conducted internally by Coloplast in an in vitro setting. Clinical experiences may vary from in vitro bench testing
1. BPS Leakage Test: Coloplast Lab Report. 2020 (tested against Dansac TRE Seal, Eakin Cohesive Seal, Hollister CeraRing, Salts Mouldable Seal, Welland Hyperseal® Washer and ConvaTec Stomadhesive® Seal). Data on fi le. VV-2832807